Rat IL1 beta CLIA Kit (RTES00012)
Rat IL1 beta (Interleukin 1 Beta) CLIA Kit
This kit uses Sandwich-CLIA as the method. The micro CLIA plate provided in this kit has been pre-coated with an antibody specific to Rat IL1 beta . Standards or samples are added to the appropriate micro CLIA plate wells and combined with the specific antibody. Then a biotinylated detection antibody specific for Rat IL1 beta and Avidin-Horseradish Peroxidase (HRP) conjugate are added to each micro plate well successively and incubated. Free components are washed away. The substrate solution is added to each well. Only those wells that contain Rat IL1 beta, biotinylated detection antibody and Avidin-HRP conjugate will appear fluorescence. The Relative light unit (RLU) value is measured spectrophotometrically by the Chemiluminescence immunoassay analyzer. The RLU value is positively associated with the concentration of Rat IL1 beta. The concentration of Rat IL1 beta in the samples can be calculated by comparing the RLU of the samples to the standard curve.
|Detection range||12.50-800 pg/mL|
|Sample type||Serum, plasma and other biological fluids|
|Repeatability||CV < 15%|
This kit recognizes Rat IL1 beta CLIA Kit in samples. No significant cross-reactivity or interference between Rat IL1 beta CLIA Kit and analogues was observed.
As the RLU values of the standard curve may vary according to the conditions of the actual assay performance (e.g. operator, pipetting technique, washing technique or temperature effects), the operator should establish a standard curve for each test. Typical standard curve and data is provided below for reference only.
Intra-assay Precision (Precision within an assay): 3 samples with low, mid range and high level Rat IL1 beta CLIA Kit were tested 20 times on one plate, respectively.
Inter-assay Precision (Precision between assays): 3 samples with low, mid range and high level Rat IL1 beta CLIA Kit were tested on 3 different plates, 20 replicates in each plate.
|Intra-assay Precision||Inter-assay Precision|
|C V (%)||10.66||10.78||9.97||11.20||10.64||8.25|
The recovery of Rat IL1 beta CLIA Kit spiked at three different levels in samples throughout the range of the assay was evaluated in various matrices.
|Sample Type||Range (%)||Average Recovery (%)|
|EDTA plasma (n=5)||100-115||107|
|Cell culture media (n=5)||94-109||102|
Samples were spiked with high concentrations of Rat IL1 beta CLIA Kit and diluted with Reference Standard & Sample Diluent to produce samples with values within the range of the assay.
|Serum (n=5)||EDTA plasma (n=5)||Cell culture media (n=5)|
Kit Components & Storage
An unopened kit can be stored at 4'C for 1 month. If the kit is not used within 1 month, store the items separately according to the following conditions once the kit is received.
|Micro CLIA Plate(Dismountable)||8 wells Ã12 strips||-20'C, 6 months|
|Reference Standard||2 vials|
|Concentrated Biotinylated Detection Ab (100Ã)||1 vial, 120 uL|
|Concentrated HRP Conjugate (100Ã)||1 vial, 120 uL||-20'C(shading light), 6 months|
|Reference Standard & Sample Diluent||1 vial, 20 mL||4'C, 6 months|
|Biotinylated Detection Ab Diluent||1 vial, 14 mL|
|HRP Conjugate Diluent||1 vial, 14 mL|
|Concentrated Wash Buffer (25Ã)||1 vial, 30 mL|
|Substrate Reagent A||1 vial, 5 mL||4'C (shading light)|
|Substrate Reagent B||1 vial, 5 mL||4'C (shading light)|
|Plate Sealer||5 pieces|
|Product Description||1 copy|
|Certificate of Analysis||1 copy|